The Sliding Hammer is a reusable instrument for the implantation or extraction of the FastLok™ Staple and Buckle system for fixation of suture and tape to cortical bone.
The instrument is manufactured to exacting tolerances of medical grade stainless steel. The Impactor Assembly is used with the Xiros Impactor.
For the implantation or extraction of the Fastlok™ Staple System
The Sliding Hammer is a stainless-steel reusable instrument designed to implant or extract the Fastlok 6 mm and 8mm Staples. The instrument consists of a stainless-steel shaft with a distal threaded end, distal impact base, sliding hammer weight and a proximal impact cap. The distal threads are inserted into the proximal end of the Impactor Assembly threaded hole. The combination of the Impactor Assembly and Sliding Hammer are an effective method to implant or extract the Fastlok Staple system.
The Sliding Hammer is intended for the insertion or extraction of the Fastlok Staple system:
- Implant or extraction of Fastlok 6 mm X 23 mm Staple
Implant or extraction of Fastlok 8 mm X 23 mm Staple
How does it work?
The Sliding Hammer is used to implant or extract the Fastlok™ 6 mm and 8 mm Staple system. The Sliding Hammer is sterilized prior to use in the surgical procedure. Once the Fastlok is secured in the adjustable jaws, the Sliding Hammer is attached to the proximal end of the Impactor Assembly by threading the distal end of the Sliding Hammer into the proximal threaded end of the Impactor Assembly. The Sliding Hammer is manipulated from proximal to distal, translating the force needed to implant the Fastlok staple. Once the staple is implanted, the sliding hammer is removed and the jaws of the Impactor Assembly are opened by rotating the proximal Impactor end counterclockwise, releasing the staple.
To extract the Fastlok staple, the Impactor Assembly is brought into the surgical field and the distal jaws are placed over the Fastlok Staple. Once attached, the Sliding Hammer is attached to the proximal end of the Impactor. The hammer is manipulated from distal to proximal, translating the force needed to remove the Fastlok staple. The Impactor Assembly, Sliding Hammer and Fastlok Staple are removed from the operative field and separated.
- 202-1118 Sliding Hammer (non-sterile)
Features and Benefits
- Stainless-steel construction
- Ergonomically shaped sliding hammer
- Threaded distal end for attachment to the Impactor Assembly
- Proximal and distal impact surfaces to translates force
- Sterilizable by multiple methods used in health care facilities
- Robust design for extended use
- Manufactured to exacting standards
Please note: Regulatory approval for products mentioned on this website varies from country to country. For further information, and for details of local distributors and agents, contact our Sales and Marketing team at email@example.com. All data listed on this page is held at Xiros and can be requested by contacting firstname.lastname@example.org.